qNano & TRPS Measurement
Control Suite Software
Automatic Fraction Collector (AFC)
The rate plot does not display a constant gradient. Why is this important and what can be done to rectify the issue?
The rate plot is used during a recording to evaluate the quality of the data. The rate plot must maintain a linear gradient, as shown below:
If there is a change to the rate of particles passing through the nanopore then this will be reflected by a change in the gradient of the rate plot. This is demonstrated in the rate plot shown below:
When this observation occurs, the blockade magnitude and particle rate measurements will be incorrect and will result in inaccurate size and concentration calculations. The system setup must be corrected:
For tips on unblocking a nanopore, read this article for advice.